For the cDAQ chassis, you do not need a kit for panel mounting. Other product and company names listed are trademarks or trade names of their respective companies. Use the following kits for each chassis:. Ring Lug Attached to Chassis Ground. With the two new chassis, many of these operations use only a single counter because of counter technology advancements, providing more available counters for additional measurements.
|Date Added:||12 April 2018|
|File Size:||47.24 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
cDAQ – National Instruments
cfaq-9174 Select this device and click Add Selected Devices. After removing the screws, replace them ni cdaq-9174 the two longer screws included in the NI desktop mounting kit. Check each module’s ni cdaq-9174 for more details. Important product and driver updates Calibration reminders to keep your hardware in compliance Notifications about warranty expiration Simplified asset management for all your products on ni.
After rebooting your computer or adding new hardware to your ni cdaq-9174, you may see the NI Product Registration Wizard appear.
NI empowers customers with off-the-shelf software such as NI LabVIEW ccdaq-9174 modular cost-effective hardware and sells xdaq-9174 a broad base of more than 30, different companies worldwide, with no one customer ni cdaq-9174 more than 3 percent of revenue and no one industry representing more than 15 percent of revenue.
Or if your chassis is automatically detected on the network, it will appear under the Ni cdaq-9174 Devices section. Ring Lug Attached to Chassis Ground.
781157-01 – National Instruments Corporation cDAQ-9174, CompactDAQ chassis (4 slot USB)
Connect Sensors or Signals Next. NI provides these screws with the panel mount kit. Headquartered in Austin, Texas, NI has more than 5, employees and direct operations in more than 40 countries. To add cdwq-9174 company click here. About National Instruments National Instruments www. With ni cdaq-9174 two new chassis, many of these operations use only a single counter because of counter ni cdaq-9174 advancements, ceaq-9174 more available counters for additional measurements.
Some counter ni cdaq-9174 on the original chassis, such as finite pulse train generation, require the use of two counters.
The screws are included in the DIN-rail kit. System Settings in MAX. You can panel mount ni cdaq-9174 with either panhead machine screws or M3. Two keyholes are located on cDAQ for mounting ni cdaq-9174 to a panel or wall. Ni cdaq-9174 screw height for both screw types is 7. Refer to the documentation included with the panel mount kit for more detailed dimensions.
Figure 12 illustrates the cdas-9174 dimensions and installation on the CompactDAQ chassis.
Use the following kits for each ni cdaq-9174.
With these external triggers, engineers can synchronise an NI CompactDAQ system with an external device or another system without installing and wiring a digital module, making synchronising systems ni cdaq-9174 and ni cdaq-9174 cost-effective.
The two new chassis build on the functionality of the original chassis by adding a four-slot option; the ability to take mixed-sensor measurements at different rates; two built-in, external BNC triggers; and four advanced counters. Ubiquity and a wide range of application support nk Windows a popular operating system for benchtop, portable laptop and production line cdsq-9174 systems. The chassis operate in a temperature range of to 55 degrees Cdq-9174 and can withstand up to 30 g shock and 3 ni cdaq-9174 vibration.
Web-Based Configuration and Monitoring. Windows is a registered trademark of Microsoft Corporation in the United States and other countries. NI Ni cdaq-9174 Mounting Kit. Alternatively, you can use the NI Network Browser located in the NI start menu folder on your host computer to ni cdaq-9174 chassis on your in subnet. The mark LabWindows is used under a license from Microsoft Corporation.